



The PLANIX EX
not only allows you to quickly and accurately measure areas,
but it also allows you to calculate lengths, coordinates,
radii, and angles.
PLANIX EXL is also available for a wider range of 450mm x
10m. 
PLANIX
EX / PLANIX 5/6/7 (3.65M)
You can obtain a free viewer by following this link

Functions 
PLANIX
EX 
PLANIX
EXL 
X, Y coordinate 
Yes 
Yes 
Side length 
Yes 
Yes 
Line 
Yes 
Yes 
Area 
Yes 
Yes 
Center
point 
Yes 
Yes 
Radius 
Yes 
Yes 
Triangle 
Yes 
Yes 
Angle 
Yes 
Yes 
RS232C cable 
Yes(Option) 
Yes(Option) 
Printer 
Yes(Option) 
Yes(Option) 

Easy Measurement
The automatic closing function, which automatically finishes
measurement when the trace point is returned to the starting
point, makes measurement easier.
Wide Variety of Measurement Units
The PLANIX EX and EXL are not only equipped with metric units
(mm, cm, m, and km) but also provide the user with such units
as in., ft., yd., mi., and acres, as well as other userdefined
units.
Two Modes Are Better Than One
With the Point mode, the user can more easily and more quickly
obtain areas and lengths of diagrams with straight lines.
The Stream mode should be used for tracing curves on curved
lines. 
RS232C interface Cable

The RS232C interface cable connects
PLANIX EX or PLANIX EXL to the computer.
Any measurement value can be sent from the PLANIX to the
computer through this cable.
It comes with a free datacollecting software. 
Thermal Printer 
There is an optional thermal printer
for an immediate hardcopy record of your measurements. 
Power 
Internal NiCd or AC adaptor 
Operating time 
40 hours after 15 hours
recharge 
Display 
LCD, 17 digits on 3 lines 
Measuring range 
EX 
380 mm x 10 m (14 in. x 32.5 ft.) 
EXL 
450 mm x 10 m (17 in. x 32.5 ft.) 
Accuracy 
+/ 0.1% (for an area
of 100 x 100mmt) 
Minimum line reading 
0.05 mm 
Dimensions 
EX 
350 x 43 x 165 mm 
EXL 
400 x 43 x 165 mm 
Carrying case 
EX 
365 x 65 x 195 mm 
EXL 
465 x 70 x 245 mm 
Weight 
EX 
Approx. 1 kg. 
EXL 
Approx. 1.4 kg. 

△=Option 
PLANIX
5 
PLANIX
6 
PLANIX
7 
PLANIX 10S
"Marble" 
PLANIX
EX 
PLANIX EXL

X,
Y coordinates 
* 
* 
* 
* 
○ 
○ 
External
outputs 
* 
* 
* 
* 
○ 
○ 
Area 
○ 
○ 
○ 
○ 
○ 
○ 
Side
length 
* 
* 
* 
○ 
○ 
○ 
Line
length 
* 
* 
* 
○ 
○ 
○ 
Radius 
* 
* 
* 
* 
○ 
○ 
Center
point 
* 
* 
* 
* 
○ 
○ 
Triangle
(base length, height and area) 
* 
* 
* 
* 
○ 
○ 
Angle
(side length and angle) 
* 
* 
* 
* 
○ 
○ 
Scale:
key input 
* 
* 
○ 
○ 
○ 
○ 
Scale:
PLOTreal measurement correction 
* 
* 
* 
○ 
○ 
○ 
Measuring
area for differing scale of X, Y 
* 
* 
Area
only 
Area
only 
○ 
○ 
Sum
and average measurement 
○ 
○ 
○ 
○ 
○ 
○ 
Decimal
point setup 
* 
* 
* 
○ 
○ 
○ 
Auto
power off 
○ 
○ 
○ 
○ 
○ 
○ 
Calculation 
* 
* 
* 
* 
○ 
○ 
Display 
dot
LCD 8 digits x 1 line 
dot
LCD 16
digits x 2 lines 
dot
LCD 17
digits x 3 lines 
Measuring
range 
300mm
x 300mm 
300mm
x 3m 
300mm
x 10m 
380mm
x 10m 
450mm
x 10m 
Minimum
line reading 
±
0.1cm 
±
0.05mm 
Accuracy 
±
0.2% 
±
0.1% 
Auto
close function 
* 
* 
* 
○ 
○ 
○ 
Automatic
numbering 
* 
* 
* 
* 
△ 
△ 
Printer 
* 
* 
* 
* 
△ 
△ 
RS232C
interface cable 
* 
* 
* 
* 
△ 
△ 

page
top 
